In the context of the urgent need for ventilators under the current Corona crisis, VUB believes it is important to make every effort to design a ventilator to help patients affected by COVID-19.
Part of this effort is to make our design files available to other groups trying to replicate our work or use our work as a bootstrap for their own development work.
VUB cannot be held liable if the hoped-for results are not achieved.
The current iteration has not been certified, or sufficiently tested to use.
It should be understood this is a work-in-progress, not a finished product.
Without the current sense of urgency we would delay publishing until sufficient testing and validation had been done.
All files are published under the under
GNU General Public License v3.0 https://choosealicense.com/licenses/gpl-3.0/
Share-and-share-alike, no closed source, commercial work allowed, attribution of authors needs to be maintained.
INSTALLATION INSTRUCTIONS (WINDOWS)
Installing the GUI is a 3-part process:
1-install the Daemon
3-Pull latest version of the software in Docker
1- INSTALLING THE DAEMON
Download csharp-Daemon from https://github.com/beademingstoestel/
-Ventilator Daemon for Windows64 can then be found under
Make a shortcut on the desktop to VentilatorDaemon.exe,
located in that map.
2- INSTALL DOCKER
After installation you will need to reboot.
3-PULL LATEST VERSION IN DOCKER
Download the (https://github.com/beademingstoestel/frontend_installer/blob/master/docker-compose-windows.yml) file and rename it to docker-compose.yml
Save this file somewhere where you can find it.
Open a terminal windows, go to the folder where you saved this file.
Run ‘docker-compose up’ in the directory containing the yml file.
Files will be downloaded and installed
3b-Updating to a new version
In the directory containing [docker-compose.yml](https://github.com/beademingstoestel/frontend_installer/blob/master/docker-compose.yml) file run following commands:
– docker-compose pull
– docker-compose up –force-recreate
Download the latest version of the daemon from [https://github.com/beademingstoestel/csharp_daemon/tree/master/VentilatorDaemon/publish](https://github.com/beademingstoestel/csharp_daemon/tree/master/VentilatorDaemon/publish).
You now have the latest version of the software.
4-RUN THE GUI
-Connect the machine / Arduino to the pc.
-Start ventilatordaemon.exe –
-Open Chrome and go to: http://localhost:3001/
The machine microcontroller (ATmega 2560) sends measured data to the pc at +20Hz, and are just sent. (no acknowledgement)
The microcontroller and the pc act as each other’s watchdog, exchanging a signal once per second. If one of both does not respond for 5 seconds, the other activates its alarm.
The ventilator will keep working when the pc fails.
The ventilator and pc (a laptop) work on separate batteries and have their own PSU.
Sensor data received by the PC (via Python) gets stored in MongoDB.
Data is processed on the pc, and displayed in the graphical user interface.
Needed operating parameters are set on the graphical user interface and sent to the arduino. This is done via a protocol where the settings are resent if no confirmation is received.
The architecture makes it possible to do remote monitoring of the machines, over LAN.
For safety reasons we do not intend to enable remote parameter change.